This article reports on a project that demonstrated a practical method for applying the concepts of signal-to-noise ratio (SNR), limits of detection, and limits of quantitation to μ-XRF generated EDS-based spectra; and it discusses the implications of such determinations, addresses spectral features that must be considered when making the calculations, and illustrates the application of these concepts to forensic examination and comparison of glass samples.
Micro X-ray fluorescence (μ-XRF) spectrometry using an energy dispersive X-ray (EDS) detector is capable of detecting certain major, minor, and trace elements that permit potential discrimination of glass fragments in forensic cases on the basis of differences in elemental composition. Often, elements used for discrimination are present at concentrations near the detection limit of the EDS system, and the decision whether to use these minor peaks in a comparative analysis has generally been left to the discretion of the examiner. The use of signal-to-noise ratios (SNRs) of spectral peaks provides additional objectivity in peak identification/label decisions and in the selection of elements in semiquantitative ratio comparisons. In addition, the use of SNRs enables calculations of limits of detection and limits of quantitation, as well as the monitoring of instrument performance. It also facilitates performance comparisons of different μ-XRF configurations. (publisher abstract modified)
- Ontogenetic patterns in frontal sinus shape: a longitudinal study using elliptical Fourier analysis
- Response to commentary on "Transfer and persistence studies of inorganic and organic gunshot residues using synthetic skin membranes"
- Optimization of InnoXtract (TM) extraction and purification system for DNA extraction from skeletal samples